Each Spotter Sensor Head is reusable and autoclavable up to 10x cycles.
- Soak sensor in disinfectant solution
- Use adequate brush or towel soaked in a disinfection solution
- Rinse well in flowing water
- Visually inspect sensor to detect possible damages to the sensor coating or to the gold plating of contacts
- pack sensor in a sterilization pouch
- Steam sterilization at 134°C (273°F) for 18 minutes in gravity type autoclave
- Store sensor in sterile packaging in a dry environment